Dental Implant Screw Remover Kit Surgical Instrument Tool Drill Driver Ratchet. Compete with NeoBiotech SR kit. The screw remover kit including. This product is a Screw Remover Kit consisting of dental implant Surgical tools Drill, Surgical. Tools, and Drivers made from medical grade materials, such as stainless steel.This product is a surgical tool developed in order safely and quickly to remove screws that. Have become fractured inside an implanted Fixture for various reasons.
Screw, a new Abutment may be connected to the Fixture. This is a surgical tool that removes a screw that has become fractured inside the Fixture. After selecting the size that is appropriate for the size of the screw that is to be removed, it. Must be used by connecting a guide to the Fixture (Conical/Internal/External Guide).It is connected to the Shank Driver, and the fractured screw can be removed by rotating. By hand in the reverse direction. If the fractured screw cannot be removed using the Claw. It must be removed using the Reverse Drill and the Screw Remover. This is a tool that creates a hole on the fractured face of the Screw that will be used by. After selecting the appropriate Reverse Drill for the size of the screw that needs to be. Removed, it must be used by connecting a guide for the fi xture Perfect/ Conical/Internal/.
The depth of the drill can be adjusted from 1 mm to 3 mm, because. The Drill body has markings. The removal of an M1.6 size Screw Should utilize the Claw.Although diff erent manufacturers produce diff erent sizes of abutment screws, except in. Rare cases, typically they utilize screws of Size M1.6, M1.8, M2.0, or M2.5.
'M' stands for a metric screw. The number indicates the external size of the screw. Example, "M1.6 Screw" indicates that it is a metric screw the external diameter of which.
This is a device that removes a fractured screw. Insert a Screw Remover that is. Appropriate for the size of the hole that has been formed using the Reverse Drill and. Rotate at a low speed (at or below 80 rpm) in the reverse direction.The Screw Remover was designed in a tapered shape so as to increase, as it turns, the. Amount of friction, which is necessary to loosen the Screw. This is a surgical tool that was designed to be connected to the Screw Remover to allow. The use of the hand, Torque Wrench, or Ratchet.
Perfect Guide & PG Hand Driver. The Perfect Guide acts as a Guide for the Reverse Drill. It is used when the screw is.
Fractured at the lower part of the female screw inside the Fixture. It is connected to PG.
Hand Driver and is directly applied to the Fixture female Screw. The Perfect Guide can be safely used because it has a stop structure inside that. Prevents the Reverse Drill from drilling more than 2 mm. They act as the Guide when using the Reverse Drill. They are used when the Perfect.
Guide cannot be used, or if the screw was fractured at a location deeper than 2 mm into. The lower part of the female screw inside the Fixture. The Guide appropriate for connecting to the Fixture must be selecte. This can be easily affi xed to the Fixture by connecting it to the 3.5 Double Hex part of the.Conical, Internal, or External Guide. The sale of this item may be subject to regulation by the U. Food and Drug Administration and state and local regulatory agencies. City State: Zhengzhou, Henan China.