Dental Implant Screw Removal Kit Surgical Tool Instrument Remover Drill Driver. Compete with NeoBiotech SR kit. Screw Remover Kit is a surgical tool developed in order safely and quickly to remove screws that have become fractured inside an implanted Fixture for various reasons. After removing the Screw, a new Abutment may be connected to the Fixture. Remove Fractured Screw, Screw Remover!No Worries About Fractured Screw. This product is a Screw Remover Kit consisting of dental implant Surgical tools Drill, Surgical. Tools, and Drivers made from medical grade materials, such as stainless steel. This is a surgical tool that removes a screw that has become fractured inside the Fixture. After selecting the size that is appropriate for the size of the screw that is to be removed, it. Must be used by connecting a guide to the Fixture (Conical/Internal/External Guide).
It is connected to the Shank Driver, and the fractured screw can be removed by rotating. By hand in the reverse direction. If the fractured screw cannot be removed using the Claw.
It must be removed using the Reverse Drill and the Screw Remover. This is a tool that creates a hole on the fractured face of the Screw that will be used by the Screw Remover. After selecting the appropriate Reverse Drill for the size of the screw that needs to be removed, it must be used by connecting a guide for the fi xture (Perfect/ Conical/Internal/External Guide). The depth of the drill can be adjusted from 1 mm to 3 mm, because the Drill body has markings.The removal of an M1.6 size Screw Should utilize the Claw. Although diff erent manufacturers produce diff erent sizes of abutment screws, except in rare cases, typically they utilize screws of Size M1.6, M1.8, M2.0, or M2.5.'M' stands for a metric screw.
The number indicates the external size of the screw. For example, "M1.6 Screw" indicates that it is a metric screw the external diameter of which is 1.6 mm. This is a device that removes a fractured screw.Insert a Screw Remover that is appropriate for the size of the hole that has been formed using the Reverse Drill and rotate at a low speed (at or below 80 rpm) in the reverse direction. The Screw Remover was designed in a tapered shape so as to increase, as it turns, the amount of friction, which is necessary to loosen the Screw. This is a surgical tool that was designed to be connected to the Screw Remover to allow. The use of the hand, Torque Wrench, or Ratchet. Perfect Guide & PG Hand Driver.
The Perfect Guide acts as a Guide for the Reverse Drill. It is used when the screw is. Fractured at the lower part of the female screw inside the Fixture. It is connected to PG. Hand Driver and is directly applied to the Fixture female Screw.
The Perfect Guide can be safely used because it has a stop structure inside that. Prevents the Reverse Drill from drilling more than 2 mm.They act as the Guide when using the Reverse Drill. They are used when the Perfect. Guide cannot be used, or if the screw was fractured at a location deeper than 2 mm into. The lower part of the female screw inside the Fixture.
The Guide appropriate for connecting to the Fixture must be selecte. This can be easily affi xed to the Fixture by connecting it to the 3.5 Double Hex part of the.
Conical, Internal, or External Guide. The sale of this item may be subject to regulation by the U. Food and Drug Administration and state and local regulatory agencies. City State: Zhengzhou, Henan China.
This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Handpieces & Instruments\Dental Instruments". The seller is "april-dent" and is located in this country: CN. This item can be shipped worldwide.